Spectral self-interference microscopy (SSM) relies on the balanced collection of light traveling two different paths from the sample to the detector, one direct and the other indirect from a reflecting substrate. The resulting spectral interference effects allow nanometer-scale axial localization of isolated emitters. To produce spectral fringes the difference between the two optical paths must be significant. Consequently, to ensure that both contributions are in focus, a low-numerical-aperture objective lens must be used, giving poor lateral resolution. Here this limitation is overcome using a 4Pi apparatus to produce the requisite two paths to the detector. The resulting instrument generalizes both SSM and 4Pi microscopy and allows a quantification of SSM resolution (rather than localization precision). Specifically, SSM is shown to be subject to the same resolution constraints as 4Pi microscopy.
© 2007 Optical Society of AmericaFull Article | PDF Article
Matthias Nagorni and Stefan W. Hell
J. Opt. Soc. Am. A 18(1) 36-48 (2001)
A. Bilenca, A. Ozcan, B. Bouma, and G. Tearney
Opt. Express 14(16) 7134-7143 (2006)
A. Bilenca, T. Lasser, A. Ozcan, R. A. Leitgeb, B. E. Bouma, and G. J. Tearney
Opt. Express 15(6) 2810-2821 (2007)