Misalignment of the aperture diaphragm present in optical microscopes results in angular illumination asymmetry (ANILAS) at the sample plane. Here we show that through-focus propagation of ANILAS results in a lateral image shift with a focus position. This could lead to substantial errors in quantitative results for optical methods that use through-focus images such as three-dimensional nanoparticle tracking, confocal microscopy, and through-focus scanning optical microscopy (TSOM). A correlation exists between ANILAS and the slant in TSOM images. Hence, the slant in the TSOM image can be used to detect, analyze, and rectify the presence of ANILAS.
© 2017 Optical Society of AmericaFull Article | PDF Article
OSA Recommended Articles
Opt. Lett. 41(4) 745-748 (2016)
Ravi Kiran Attota, Peter Weck, John A. Kramar, Benjamin Bunday, and Victor Vartanian
Opt. Express 24(15) 16574-16585 (2016)
Ravi Kiran Attota
Opt. Express 24(20) 22616-22627 (2016)