Rotational-level-dependent dephasing rates and nonresonant background can lead to significant uncertainties in coherent anti-Stokes Raman scattering (CARS) thermometry under high-pressure, low-temperature conditions if the gas composition is unknown. Hybrid femtosecond/picosecond rotational CARS is employed to minimize or eliminate the influence of collisions and nonresonant background for accurate, frequency-domain thermometry at elevated pressure. The ability to ignore these interferences and achieve thermometric errors of <5% is demonstrated for N2 and O2 at pressures up to 15 atm. Beyond 15 atm, the effects of collisions cannot be ignored but can be minimized using a short probe delay (~6.5 ps) after Raman excitation, thereby improving thermometric accuracy with a time- and frequency-resolved theoretical model.
© 2012 OSA
Coherent anti-Stokes Raman scattering (CARS) is a highly accurate and precise nonlinear spectroscopic technique that has been utilized for the measurement of gas-phase temperature in a wide range of reacting and non-reacting flows . While CARS offers excellent chemical specificity and efficient signal collection, measurement errors from composition-dependent interferences, such as collisional energy transfer and nonresonant background, significantly complicate the analysis, particularly in high-pressure environments . As shown in previous work, interferences from nonresonant background can be suppressed in the frequency domain using polarization techniques  or by temporally separating the probe pulse from the pump and Stokes pulses [4–6]. Temporal suppression has been demonstrated for both vibrational  and rotational CARS , but the temporal probe delays in previous work were sufficiently long (~100 ps) that the rovibrational and rotational spectra were sensitive to the effects of collisions on individual rotational transitions, especially at low temperature. In fact, much of the early work in time-domain fs and ps CARS utilized this sensitivity to measure molecular and collision-dependent parameters . This included the effects of collisions on Q-branch line-shift parameters , rotational constants , and rotational energy transfer models [12–14] for several molecules of interest in combustion (H2, N2, CO, and C2H2). Such measurements were intended to provide critical information for modeling the composition-dependent line widths of gas-phase spectra for accurate thermometry.
When the gas composition is not known a priori, however, it is advantageous to avoid entirely the effects of collisions. Whereas previous ultrafast CARS temperature measurements exploited features of the molecular response at long delay times and, therefore, required detailed modeling of collisional effects , more recent efforts have instead utilized the initial (<15 ps) decay rate of the Raman excitation, which occurs through non-collisional, frequency-spread dephasing . Frequency-spread dephasing of the vibrational manifold is much faster than collisional dephasing for pressures up to 10 atm ; therefore, uncertainties due to the complex collisional environment can be avoided. However, single-shot detection in the time domain requires chirped-probe detection [1, 15], and interference-free measurements beyond 1 atm have yet to be demonstrated using this approach. Hence it is not possible at this time to ascertain the accuracy of interference-free, chirped-probe fs CARS at elevated pressures.
As an alternative to time-domain detection, it is of interest to determine if measurements that are free of collisions and nonresonant-background can be achieved using frequency-domain thermometry at high pressure. Recently, the authors demonstrated a hybrid fs/ps CARS technique that employed 100-fs pump and Stokes pulses to induce coherent vibrational or pure rotational Raman oscillations in N2 while probing the molecular response using a frequency-narrowed, time-delayed 1–10 ps probe pulse [17–20]. In this approach, a short (0.3–10 ps) probe-pulse delay was sufficient to avoid overlap with the pump and Stokes pulses to eliminate nonresonant background, while the transform-limited ps pulse allowed for single-shot frequency-domain detection of individual N2 rotational  or vibrational transitions . In the current work, we investigate the ability of hybrid fs/ps rotational coherent anti-Stokes Raman scattering (RCARS) to avoid the effects of collisions and nonresonant background for accurate, interference-free thermometry at high pressure. The ability to avoid collisional effects at high pressure is of particular interest for frequency-domain thermometry, which typically carries inherent sensitivity to rotational-level-dependent linewidths in the case of ns or ps CARS. The ability of the current approach to isolate individual rotational transitions enables detailed analysis of the effects of rotational-level-dependent dephasing rates on thermometric accuracy. Expanding the technique to include fs/ps RCARS thermometry of O2, we use a phenomenological model to compare the relative effects of collisions on N2 and O2 spectra for probe delays of 6.5 to 150 ps at elevated pressure. For conditions in which collisions cannot be avoided (beyond 15 atm), we evaluate the feasibility of improving temperature accuracy by minimizing the effects of collisions with short probe delays and including the collisional linewidth in a time- and frequency-resolved theoretical model.
2. Theory and experiment
A detailed theoretical treatment of hybrid fs/ps CARS and experimental description can be found elsewhere [17, 21]. For pure S-branch transitions (Δv = 0, ΔJ = + 2), the molecular response from the pump (ωpump) and Stokes (ωStokes) pulses is treated phenomenologically as a function of time, t, in Eq. (1),17, 21]. The transition intensity IJ + 2,J is not a function of time, but is highly dependent on temperature, T. The response decays exponentially with a time constant τJ + 2,J = ΓJ + 2,J−1. The linewidth is highly temperature and pressure dependent, displaying a linear dependence on pressure and more complex inverse relationship with temperature . Because low rotational states decay more rapidly than high rotational states, the spectral and temporal features of the CARS signal begin to mimic higher temperatures at long delay times [8, 17, 23]. The resulting thermometric error is more pronounced at low temperature where the difference between linewidths of low and high rotational states is largest [8, 17]. At higher pressure, rapid collisional dephasing can also significantly enhance rotational energy transfer and require detection at short delay times for accurate interference-free thermometry.
N2 and O2 CARS spectra were recorded at 298 and 295 K, respectively, from 1 to 20 atm. The pure gases were pressurized in a cylindrical stainless-steel vessel, and the temperature was measured with a 1/16-inch diameter K-type thermocouple and pressure with a digital pressure gauge (0–500 ± 1.25 psi). The frequency-dispersed RCARS signal is detected using a 0.303-m spectrometer (1200 line/mm) with sufficient resolution for distinguishing individual rotational transitions. The signal can be numerically integrated over the entire S-branch (~10–200 cm−1) as given in Fig. 1 for N2 and O2 at pressures of 1, 2.5, 5, 10, 15, and 20 atm, or numerically integrated over a single transition.
The solid lines in Fig. 1 represent simulations using a time- and frequency-domain theoretical model with modified exponential gap (MEG) linewidth parameters from Seeger et al.  and S-branch linewidths approximated following Martinsson et al. . The individual linewidths vary as a function of initial rotational state, J, decreasing as J is increased. Thus the low rotational states exhibit higher rates of energy transfer and decay at a faster rate than high rotational states. As highlighted in our previous publications [17,20], the linewidth varies from 0.098 cm−1 to 0.063 cm−1 from J = 4 to 18 for N2, and 0.107 cm−1 to 0.064 cm−1 for N = 5 to 23 for O2. Since each transition decays at a different rate, the RCARS spectra appear to shift to higher temperatures as the low rotational states decay rapidly with increased probe delay. The experimentally measured time decays are well predicted by the theoretical model in Fig. 1 and illustrate the short time window between nonresonant background and strong collisional dephasing at high pressure. The dephasing rate for N2 is faster than that of O2, which implies that N2 CARS thermometry will be more sensitive to collisions.
3. Results and discussion
Because of rapid collisional dephasing at high pressure, the probe-pulse spectral bandwidth was increased from previous work (2.1 cm−1)  to 3.6 cm−1 (4.1 ps) to achieve a minimum probe delay of 6.5 ps (~1000 × nonresonant suppression). This bandwidth is sufficient to avoid spectral overlap of the transitions so that the effects of rotational energy transfer can be clearly observed in the frequency-domain spectra. For illustration, experimental O2 spectra are presented in Fig. 2 at fixed delays of (a) 6.5 and (b) 25 ps and pressures of 1 atm and 10 atm (symbols). The theoretical O2 spectrum (solid lines) assumes a pressure of 1 atm. At a probe delay of 6.5 ps, shown in Fig. 2(a), the difference between the two experimental spectra is negligible, leading to best-fit temperatures that are within 4% of 295 K. This illustrates the independence from collisional effects at short probe delays. For the longer probe delay of 25 ps, as shown in Fig. 2(b), the experimental spectrum at 1 atm is nearly the same as that shown at 6.5 ps. However, for a probe delay of 25 ps and a pressure of 10 atm, the spectrum shows an apparent population shift towards higher rotational states, leading to a best-fit temperature of 342 K. This represents an error of 16%.
As noted earlier avoiding the effects of collisions is important when the gas composition is not known a priori, but at elevated pressure this requires that the probe pulse be shorter than and arrive sooner than 25 ps. To evaluate the potential for interference-free measurements from 1 to 20 atm, the thermometric error was investigated from a minimum probe delay of 6.5 ps (to avoid nonresonant background) to a maximum of 150 ps (typical of ps CARS with probe pulses of ~80 ps or higher). This investigation was performed by fitting the time-dependent spectra using a spectral database where the linewidth was set to zero in the molecular response function (Eq. (1)) for all rotational transitions. Best-fit temperatures are plotted as a function of probe delay in Figs. 3(a) and 4(a) . The constant solid line represents the actual temperature, and the constant dashed lines represent ± 5% error.
The data of Figs. 3(a) and 4(a) show the quantitative impact of collisions on thermometric error associated with N2 or O2 CARS spectra at elevated pressure. At the shortest probe delay of 6.5 ps, for example, the thermometric error without corrections for nonresonant background or collisional energy transfer is <5% up to 15 atm. This error is slightly higher than that previously reported using validated RCARS models for high resolution frequency domain thermometry at elevated pressure [2, 25]. However, it should be noted that the measurements presented here differ substantially from previous frequency-domain experiments since the composition-dependent Raman linewidth component of the molecularresponse function is neglected. Hence, sensitivity to linewidth variations due to changes in gas composition would also be reduced or eliminated. This is an important finding because corrections for linewidth variations increase model complexity and are difficult to approximate in gas mixtures with unknown composition. Moreover, the error associated with collisional effects is further reduced at higher temperature [8, 17] such that interference-free measurements may be possible beyond 15 atm in heated flows.
Near room temperature, the interference-free assumption is limited even for probe delays as short as 6.5 ps where the error increases to 9% at 20 atm, thus requiring corrections for the effects of collisions. In cases for which corrections are necessary, however, the current data indicate that the use of short probe delays significantly reduces the impact of collisions and minimizes the corresponding corrections for rotational-level-dependent dephasing with the time- and frequency-resolved theoretical model. As shown in Figs. 3(b) and 4(b), the theoretical simulation with linewidths from the MEG model predicts temperatures that are within 5% of the known temperatures for all pressures and probe delays. At shorter probe delays (<100 ps), the accuracy is improved to 2.5% for all pressures, and at the shortest probe delay (6.5 ps), the accuracy improves to 1% for all pressures. The fact that the theoretical fit is improved at shorter probe delays is indicative of reduced interference from collisional effects, which could be critical in environments with unknown species compositions. Since the composition is known in this case, the increase in thermometric error at 150 ps may be partially attributed to uncertainties in linewidths derived from the MEG model, as well as uncertainties due to the Q-branch approximation. By minimizing the necessary corrections, the fs/ps hybrid RCARS approach reduces the effects of errors associated with uncertainties in linewidth.
To characterize further the sensitivity to collisional dephasing and predict thermometric errors for arbitrary pressures and probe delays, we utilized a phenomenological model, Eq. (2), that captures the increase in thermometric error with pressure and time (as indicated by the solid lines in Figs. 3 and 4),Eq. (2). This is due to greater uniformity in linewidth across the rotational energy levels of O2. This simple model can also be used to estimate the probe delay, τ23, and the associated minimum probe pulse width necessary for avoiding the effects of collisions at even higher pressures with frequency-domain ps or fs/ps CARS thermometry. At 20 atm, for example, the temperature shift is as high as 14.5 K/ps for N2 and 5.7 K/ps for O2. Without this knowledge, the use of longer probe delays or slight jitter between the preparation and probe pulses can cause significant variations in the apparent temperature (as high as 5% per ps).
In summary, we have quantified the effects of pressure on hybrid fs/ps CARS thermometric errors for both N2 and O2 up to 20 atm. Utilizing fs pump and Stokes pulses and a probe-pulse delay of 6.5 ps, it is possible to avoid the effects of collisions for pressures up to 15 atm and minimize corrections for collisional effects at even higher pressures. This can be achieved while avoiding nonresonant background and isolating individual rotational transitions for accurate frequency-domain thermometry. Future work will focus on testing the interference-free assumption at higher temperatures and pressures.
Funding was provided, in part, by the National Science Foundation (CBET-1056006, Dr. A. Atreya, Program Official) and Air Force Office of Scientific Research (Drs. T. Curcic and M. Birkan, Program Managers). J. Miller was supported by the National Science Foundation Graduate Fellowship Program. The authors also thank M. Johnson and B. Halls of Iowa State University and Drs. S. Danczyk and D. Talley of the Air Force Research Laboratory.
References and links
1. S. Roy, J. R. Gord, and A. K. Patnaik, “Recent advances in coherent anti-Stokes Raman scattering spectroscopy: Fundamental developments and applications in reacting flows,” Prog. Energ. Combust. Sci. 36(2), 280–306 (2010). [CrossRef]
2. T. Seeger, F. Beyrau, A. Brauer, and A. Leipertz, “High-pressure pure rotational CARS: comparison of temperature measurements with O2, N2 and synthetic air,” J. Raman Spectrosc. 34, 932–939 (2003). [CrossRef]
3. F. Vestin, M. Afzelius, and P. E. Bengtsson, “Development of rotational CARS for combustion diagnostics using a polarization approach,” Proc. Combust. Inst. 31(1), 833–840 (2007). [CrossRef]
5. H. M. Frey, P. Beaud, T. Gerber, B. Mischler, P. Radi, and A. P. Tzannis, “Femtosecond nonresonant degenerate four-wave mixing at atmospheric pressure and in a free jet,” Appl. Phys. B 68(4), 735–739 (1999). [CrossRef]
6. P. Beaud, H. M. Frey, T. Lang, and M. Motzkus, “Flame thermometry by femtosecond CARS,” Chem. Phys. Lett. 344(3-4), 407–412 (2001). [CrossRef]
7. T. R. Meyer, S. Roy, and J. R. Gord, “Improving signal-to-interference ratio in rich hydrocarbon-air flames using picosecond coherent anti-Stokes Raman scattering,” Appl. Spectrosc. 61(11), 1135–1140 (2007). [CrossRef] [PubMed]
8. T. Seeger, J. Kiefer, A. Leipertz, B. D. Patterson, C. J. Kliewer, and T. B. Settersten, “Picosecond time-resolved pure-rotational coherent anti-Stokes Raman spectroscopy for N(2) thermometry,” Opt. Lett. 34(23), 3755–3757 (2009). [CrossRef] [PubMed]
9. G. Knopp, P. Beaud, P. Radi, M. Tulej, B. Bougie, D. Cannavo, and T. Gerber, “Pressure-dependent N2 Q-branch fs-CARS measurements,” J. Raman Spectrosc. 33(11-12), 861–865 (2002). [CrossRef]
10. T. Lang and M. Motzkus, “Determination of line shift coefficients with femtosecond time resolved CARS,” J. Raman Spectrosc. 31(1-2), 65–70 (2000). [CrossRef]
11. T. Lang, M. Motzkus, H. M. Frey, and P. Beaud, “High resolution femtosecond coherent anti-Stokes Raman scattering: Determination of rotational constants, molecular anharmonicity, collisional line shifts, and temperature,” J. Chem. Phys. 115(12), 5418–5426 (2001). [CrossRef]
12. P. Beaud and G. Knopp, “Scaling rotationally inelastic collisions with an effective angular momentum parameter,” Chem. Phys. Lett. 371(1-2), 194–201 (2003). [CrossRef]
13. P. Beaud, T. Gerber, P. Radi, M. Tulej, and G. Knopp, “Rotationally inelastic collisions between N2 and rare gases: An extension of the angular momentum scaling law,” Chem. Phys. Lett. 373(3-4), 251–257 (2003). [CrossRef]
14. G. Knopp, P. Radi, M. Tulej, T. Gerber, and P. Beaud, “Collision induced rotational energy transfer probed by time-resolved coherent anti-Stokes Raman scattering,” J. Chem. Phys. 118(18), 8223–8233 (2003). [CrossRef]
15. T. Lang and M. Motzkus, “Single-shot femtosecond coherent anti-Stokes Raman scattering thermometry,” J. Opt. Soc. Am. B 19(2), 340–344 (2002). [CrossRef]
16. R. P. Lucht, S. Roy, T. R. Meyer, and J. R. Gord, “Femtosecond coherent anti-Stokes Raman scattering measurement of gas temperatures from frequency-spread dephasing of the Raman coherence,” Appl. Phys. Lett. 89(25), 251112 (2006). [CrossRef]
17. J. D. Miller, S. Roy, M. N. Slipchenko, J. R. Gord, and T. R. Meyer, “Single-shot gas-phase thermometry using pure-rotational hybrid femtosecond/picosecond coherent anti-Stokes Raman scattering,” Opt. Express 19(16), 15627–15640 (2011). [CrossRef] [PubMed]
18. J. D. Miller, M. N. Slipchenko, T. R. Meyer, H. U. Stauffer, and J. R. Gord, “Hybrid femtosecond/picosecond coherent anti-Stokes Raman scattering for high-speed gas-phase thermometry,” Opt. Lett. 35(14), 2430–2432 (2010). [CrossRef] [PubMed]
19. J. D. Miller, M. N. Slipchenko, and T. R. Meyer, “Probe-pulse optimization for nonresonant suppression in hybrid fs/ps coherent anti-Stokes Raman scattering at high temperature,” Opt. Express 19(14), 13326–13333 (2011). [CrossRef] [PubMed]
20. J. D. Miller, S. Roy, J. R. Gord, and T. R. Meyer, “Communication: Time-domain measurement of high-pressure N2 and O2 self-broadened linewidths using hybrid femtosecond∕picosecond coherent anti-Stokes Raman scattering,” J. Chem. Phys. 135(20), 201104 (2011). [CrossRef] [PubMed]
21. B. D. Prince, A. Chakraborty, B. M. Prince, and H. U. Stauffer, “Development of simultaneous frequency- and time-resolved coherent anti-Stokes Raman scattering for ultrafast detection of molecular Raman spectra,” J. Chem. Phys. 125(4), 044502 (2006). [CrossRef] [PubMed]
22. L. A. Rahn and R. E. Palmer, “Studies of nitrogen self-broadening at high-temperature with inverse Raman-spectroscopy,” J. Opt. Soc. Am. B 3(9), 1164–1169 (1986). [CrossRef]
23. B. Lavorel, H. Tran, E. Hertz, O. Faucher, P. Joubert, M. Motzkus, T. Buckup, T. Lang, H. Skenderovi, G. Knopp, P. Beaud, and H. M. Frey, “Femtosecond Raman time-resolved molecular spectroscopy,” C. R. Phys. 5, 215–229 (2004). [CrossRef]
24. L. Martinsson, P. E. Bengtsson, M. Alden, S. Kroll, and J. Bonamy, “A test of different rotational Raman linewidth models: accuracy of rotational coherent anti-Stokes-Raman scattering thermometry in nitrogen from 295 to 1850 K,” J. Chem. Phys. 99(4), 2466–2477 (1993). [CrossRef]
25. M. Afzelius, P. E. Bengtsson, J. Bood, J. Bonamy, F. Chaussard, H. Berger, and T. Dreier, “Dual-broadband rotational CARS modelling of nitrogen at pressures up to 9 MPa II. Rotaitonal Raman linewidths,” Appl. Phys. B 75(6-7), 771–778 (2002). [CrossRef]