The multi-pass solution for surface measurements with the help of a Shack-Hartmann sensor (SHS) on the basis of a Fizeau cavity enables fast access to surface deviation data due to the high speed of the SHS and easy referencing of the measured data through difference measurements. The multi-pass solution described in a previous publication [J. Schwider, Opt. Express 16, 362 (2008)], provides highly sensitive measurements of small displacements caused by thermal non-equilibrium states of the test set up. Here, we want to demonstrate how a pulsed thermal load changes the surface geometry. In addition the temporal response for different plate materials is monitored through a fast wave front measurement with very high sensitivity. The thermal load close to a delta-function in time will be applied from the back-side of a plane plate by heating a small Peltier element with a heat impulse of known order of magnitude. The development of the surface deviation on the time axis can be monitored by storing a set of successive deviation pictures.
© 2010 OSAFull Article | PDF Article
J. Pfund, N. Lindlein, J. Schwider, R. Burow, Th. Blümel, and K.-E. Elssner
Opt. Lett. 23(10) 742-744 (1998)
Appl. Opt. 36(7) 1433-1437 (1997)
Alexander Chernyshov, Uwe Sterr, Fritz Riehle, Jürgen Helmcke, and Johannes Pfund
Appl. Opt. 44(30) 6419-6425 (2005)