We present a novel method to accurately measure 3D polishing influence functions by using a swing arm profilometer (SAP) and a laser tracker. The laser tracker is used to align the SAP and measure the parameters of the SAP setup before measuring the influence function. The instruments and the measurement method are described, together with measurement uncertainty analysis. An influence function deliberately produced with asymmetric form in order to create a challenging test is measured, and compared with that of a commercial 3D profilometer. The SAP result is 48.2μm in PV, 7.271mm3 in volume. The 3D profilometer result is 48.4μm in PV, 7.289mm3 in volume. The forms of the two results show excellent correlation. This gives confidence of the viability of the SAP method for larger influence functions out of range of the commercial instrument.
©2010 Optical Society of America
OSA participates in Crossref's Cited-By Linking service. Citing articles from OSA journals and other participating publishers are listed here.
Alert me when this article is cited.
Equations on this page are rendered with MathJax. Learn more.