The problem of obtaining accurate photometric data at brightness levels below 0.5 foot-lambert is reviewed briefly. A method of calibrating a photoelectric photometer for heterochromatic measurements in this so-called Purkinje range is described with some details given of the design of the instrument used. By arbitrarily dividing the luminous emission of a tungsten lamp into twelve overlapping spectral bands it has been possible to calibrate a photometer, using a single correction filter for the phototube. The practical advantages of such a calibration include much time saved in brightness measurements on phosphor samples of different colors.
© 1952 Optical Society of AmericaFull Article | PDF Article
Ch. Fabry and Karl K. Darrow
J. Opt. Soc. Am. 10(5) 521-547 (1925)
Clayton H. Sharp, S. McK. Gray, W. F. Little, and H. J. Eckweiler
J. Opt. Soc. Am. 23(7) 234-245 (1933)
Kasson S. Gibson
J. Opt. Soc. Am. 30(2) 51-61 (1940)