The paper describes an interference refractometer for liquids and gases which operates automatically and reads out in digital or analog form. A compensating technique using white light is used for measurement. Zero adjustment is achieved by rotating the compensator and capturing the zero-order white light fringe by photoelectric means. Measurement of the path difference compensated by the compensator is based on electronic interpolation and counting of interference fringes by optointerferometric means, a time division multiplex technique with pulse amplitude modulation being used to obtain the electrical fringe signals.
© 1968 Optical Society of AmericaFull Article | PDF Article
OSA Recommended Articles
George W. Taylor
Appl. Opt. 12(6) 1227-1239 (1973)
J. Opt. Soc. Am. 47(6) 536-544 (1957)
A. A. Balashov, V. S. Bukreev, N. G. Kultepin, I. N. Nesteruk, E. B. Perminov, V. A. Vagin, and G. N. Zhizhin
Appl. Opt. 17(11) 1716-1722 (1978)