This work introduces a fast, low-cost, robust method based on fringe pattern and phase shifting to obtain three-dimensional (3D) mouse surface geometry for fluorescence molecular tomography (FMT) imaging. We used two pico projector/webcam pairs to project and capture fringe patterns from different views. We first calibrated the pico projectors and the webcams to obtain their system parameters. Each pico projector/webcam pair had its own coordinate system. We used a cylindrical calibration bar to calculate the transformation matrix between these two coordinate systems. After that, the pico projectors projected nine fringe patterns with a phase-shifting step of onto the surface of a mouse-shaped phantom. The deformed fringe patterns were captured by the corresponding webcam respectively, and then were used to construct two phase maps, which were further converted to two 3D surfaces composed of scattered points. The two 3D point clouds were further merged into one with the transformation matrix. The surface extraction process took less than 30 seconds. Finally, we applied the Digiwarp method to warp a standard Digimouse into the measured surface. The proposed method can reconstruct the surface of a mouse-sized object with an accuracy of 0.5 mm, which we believe is sufficient to obtain a finite element mesh for FMT imaging. We performed an FMT experiment using a mouse-shaped phantom with one embedded fluorescence capillary target. With the warped finite element mesh, we successfully reconstructed the target, which validated our surface extraction approach.
© 2015 Optical Society of AmericaFull Article | PDF Article
9 November 2015: A correction was made to Figure 14.
OSA Recommended Articles
Haosong Yue, Yue Yu, Weihai Chen, and Xingming Wu
Opt. Express 26(22) 28544-28559 (2018)
Xuexing Li, Zhijiang Zhang, and Chen Yang
Appl. Opt. 55(34) 9895-9906 (2016)
Hector R.A. Basevi, James A. Guggenheim, Hamid Dehghani, and Iain B. Styles
Opt. Express 21(6) 7222-7239 (2013)