A refractive index sensor based on the thinned and microstructure fiber Bragg grating (ThMs-FBG) was proposed and realized as a chemical sensing. The numerical simulation for the reflectance spectrum of the ThMs-FBG was calculated and the phase shift down-peak could be observed from the reflectance spectrum. Many factors influencing the reflectance spectrum were considered in detail for simulation, including the etched depth, length, and position. The sandwich-solution etching method was utilized to realize the microstructure of the ThMs-FBG, and the photographs of the microstructure were obtained. Experimental results demonstrated that the reflectance spectrum, phase shift down-peak wavelength, and reflected optical intensity of the ThMs-FBG all depended on the surrounding refractive index. However, only the down-peak wavelength of the ThMs-FBG changed with the surrounding temperature. Under the condition that the length and cladding diameter of the ThMs-FBG microstructure were 800 and , respectively, and the position of the microstructure of the ThMs-FBG is in the middle of grating region, the refractive index sensitivity of the ThMs-FBG was ∕refractive index unit with the wide range of 1.33–1.457 and a high resolution of . The temperature sensitivity was , which was approximately equal to that of common FBG.
© 2008 Optical Society of AmericaFull Article | PDF Article
Xianfeng Chen, Kaiming Zhou, Lin Zhang, and Ian Bennion
Appl. Opt. 44(2) 178-182 (2005)
Domenico Paladino, Agostino Iadicicco, Stefania Campopiano, and Andrea Cusano
Opt. Express 17(2) 1042-1054 (2009)
Christophe Caucheteur, Patrice Mégret, and Andrea Cusano
Appl. Opt. 48(20) 3915-3920 (2009)