We developed a new, to the best of our knowledge, test method to measure the wavefront error of the high-NA optics that is used to read the information on the high-capacity optical data storage devices. The main components are a pinhole point source and a Shack–Hartmann sensor. A pinhole generates the high-NA reference spherical wave, and a Shack–Hartmann sensor constructs the wavefront error of the target optics. Due to simplicity of the setup, it is easy to use several different wavelengths without significant changes of the optical elements in the test setup. To reduce the systematic errors in the system, a simple calibration method was developed. In this manner, we could measure the wavefront error of the NA 0.9 objective with the repeatability of 0.003λ rms and the accuracy of 0.01λ rms.
© 2007 Optical Society of AmericaFull Article | PDF Article
Lin Li, Stanislaw Szapiel, and Claude Delisle
Appl. Opt. 31(7) 866-875 (1992)
Carolyn R. Mercer and Katherine Creath
Appl. Opt. 35(10) 1633-1642 (1996)
Yucong Zhu, Katsumi Sugisaki, Masashi Okada, Katsura Otaki, Zhiqiang Liu, Jun Kawakami, Mikihiko Ishii, Jun Saito, Katsuhiko Murakami, Masanobu Hasegawa, Chidane Ouchi, Seima Kato, Takayuki Hasegawa, Akiyoshi Suzuki, Hideo Yokota, and Masahito Niibe
Appl. Opt. 46(27) 6783-6792 (2007)